High Quality for Cable Smoke Density Test Apparatus - ISO 8124-4 Alu alloy Impact Head from Swing Elements without Accelerometer – Skyline
High Quality for Cable Smoke Density Test Apparatus - ISO 8124-4 Alu alloy Impact Head from Swing Elements without Accelerometer – Skyline Detail:
ISO 8124-4 Impact Head from swing elements (without Accelerometer)
Product information of the Impact Head For Swing
Test mass, consisting of an aluminium sphere or semi-sphere of radius 80 mm ± 3 mm, and a total mass (including accelerometer) of 4,6 kg ± 0,05 kg. The impacting part between the surface struck and the accelerometer shall be homogeneous and free from voids. Cables connected to the accelerometer shall be placed in such a way that the effect on the mass of the test mass is minimized
Test Standard: ISO8124-4 6.4.1
Assemble and install the swing element to be tested in accordance with the manufacturer’s instructions. Suspend the swing with the means of suspension that has been supplied with the swing and at the maximum height that these permit. If ropes or cables are the means of suspension, they may need to be stretched in order to allow smooth travel when the swing is released during the test. If needed, apply a load of, for example, 5 kg to the end of each rope or cable and leave it for 6 h or until the ropes or cables have been straightened. Adjust all parts of the set-up so that the suspending chains for the test mass are parallel to the means of suspension for the swing element. Suspend and adjust the test mass so that the contact point of the swing element and the centre of the test mass are in the same horizontal plane as the centre of gravity of the test mass. Ensure that the chains for test mass are not twisted and that the test mass hangs in a vertical line. Affix an index mark to the side of swing elements that are supported by chains, ropes, cables or other non-rigid suspending elements. The index mark may be on any part of the suspended member that is immediately below the pivot point in the free-hanging rest position. Swing elements that are supported by chains, ropes, cables or other non-rigid suspending elements shall be raised along their arc of travel until the side-view projection of a straight line through the pivot point and index mark forms an angle of 60° ± 1° with the vertical. Once the suspended member is raised to the test position, some curvature will be produced in the suspending elements. Adjust the suspended member position to determine that curvature which provides a stable trajectory.
Product detail pictures:
Related Product Guide:
Attaining consumer satisfaction is our firm's purpose for good. We'll make wonderful efforts to produce new and top-quality merchandise, meet up with your special necessities and supply you with pre-sale, on-sale and after-sale products and services for High Quality for Cable Smoke Density Test Apparatus - ISO 8124-4 Alu alloy Impact Head from Swing Elements without Accelerometer – Skyline , The product will supply to all over the world, such as: Switzerland , Philippines , Greek , Wide selection and fast delivery to suit your needs! Our philosophy: Good quality, great service, keep improving. We've been looking forward that more and more oversea friends join in our family for further development near the future!
By Gill from Durban - 2017.10.25 15:53
This manufacturer can keep improving and perfecting products and service, it is in line with the rules of market competition, a competitive company.
By Helen from Kuwait - 2018.12.10 19:03