Factory Cheap Cables Smoke Density Test Room - Contact Heat Transimmision Test – Skyline
Factory Cheap Cables Smoke Density Test Room - Contact Heat Transimmision Test – Skyline Detail:
Contact Heat Transimmision Test Apparatus Room Temperature to 500 Temperature Automatic Programming Control
Product Profile of the Contact Heat Transimmision Test
Contact heat resistance test device, developed according to EN702 test standard, also meets the requirements of domestic AQ 6103-2007 welding protective gloves contact heat resistance test standard. The test results can show the contact temperature, the starting time and the threshold time. The principle is that under the set contact temperature, the protective fabric can contact the heated cylinder at the same time withstanding the pressure of 49N. The time when the temperature rises from the initial temperature to 10 degrees C is called the threshold time, and the time is used to evaluate the protective weaving. Thermal protection performance of the product.
Product feature of the Contact Heat Transimmision Test
- Metal heater with high temperature resistant material was used, and the diameter of contact surface was 25.2±0.05 mm.
- Heater heating range: room temperature to 500 C, man-machine interface can be real-time display;
- The contact surface of calorimeter 25±0.05×D5±0.02mm, and the range of temperature measurement is not less than 500 C.
- Contact force: 49N, contact speed: 5mm/s, calorimeter servo motor control, can automatically rise and fall, program control can automatically complete the entire test process;
- The heater platform can move left and right in parallel, which is convenient for loading and can ensure the safety of operators.
- LCD display panel, PLC automatically records the start temperature, threshold time, threshold time termination temperature and real-time temperature.
Product parameters of the Contact Heat Transimmision Test
|Standards||EN 702 、ISO 12127-1|
|The diameter of the contact surface||￠25.2±0.05 mm|
|Heater Heating Range||Room Temperature to 500 Temperature|
|Temperature measurement range||≥ 500 C|
|Critical time timer||0-99.99 seconds, accuracy: +0.01 seconds|
|Sample size||80 mm in diameter|
|Weight||600(D) X 250(mm) X 650(mm)|
|Gas Supplies||Propane and air compressed air|
Contact Heat Transimmision Test Apparatus is applicable to protective clothing (including hand protectors) or its constituent materials intended to protect against high contact temperature. The application of the standard is restricted to contact temperatures from 100 °C to 500 °C.
The apparatus is designed and manufactured according to EN 702 and ISO 12127-1 standards.
Product detail pictures:
Related Product Guide:
"Control the quality by the details, show the strength by quality". Our company has strived to establish a highly efficient and stable staff team and explored an effective quality control process for Factory Cheap Cables Smoke Density Test Room - Contact Heat Transimmision Test – Skyline , The product will supply to all over the world, such as: London , luzern , Montpellier , To have much more enterprise. ompanions, we've got updated the item list and seek for optimistic co-operation. Our web-site shows the latest and complete information and facts about our goods list and company. For further acknowledge, our consultant service group in Bulgaria will reply to all of the inquiries and complications immediately. They're going to make their finest effort to meet buyers need. Also we support the delivery of absolutely free samples. Business visits to our business in Bulgaria and factory are generally welcome for a win-win negotiation. Hope to expertise a happy company co-operation perform with you.
By Modesty from South Africa - 2018.06.09 12:42
As an international trading company, we have numerous partners, but about your company, I just want to say, you are really good, wide range, good quality, reasonable prices, warm and thoughtful service, advanced technology and equipment and workers have professional training, feedback and product update is timely, in short, this is a very pleasant cooperation, and we look forward to the next cooperation!
By Nicole from Italy - 2018.06.18 19:26